The MA919B/1 probe is designed for use with the Milwaukee Mi456 Total Acidity Mini Titrators Meters.
Top features include:
Glass body is ideal use in wine applications as it is easily cleaned and gives fast results.
The double junction means that virtually no silver from the reference cell is mixed with the sample and reduces the risks of clogging.
The ability to refill the electrolyte solution extends the life of the probe. Use MA9011 3.5M KCl fill solution. With a refillable pH electrode, best results are achieved if the fill cap is removed to create a higher flow rate. This results in a faster and more stable reading.
Care and Use
Properly maintaining your pH probe will give you fast, accurate results for longer.
Here are our top four tips.
1. Always store with the cap full of buffer or storage solution not water.
2. Clean your probe regularly by soaking in cleaning solution.
3. Calibrate after cleaning and/or storage before testing to adjust for changes in the pH probe sensor.
4. Gently stir the probe in the sample for a faster result.
Reference: Double, Ag/AgCl
Junction: Ceramic, Single
Electrolyte: KCl 3.5M
Range: 0-14 pH
Tip Shape: Sphere
Diameter: 0.5 inches
Dimensions: 4.7 inches / 6.4 inches
Operating Temperature: -5 to 70°C (23 to 158°F)
Cable length: 3.3 feet